There are currently no product reviews.
The AKAI 1720 model reel to reel tape recorder described in this Manual is quite an old unit - circa late 1960's. As a consequence, the description of the mechanical details - and adjustments thereof - is quite critical. The manual does this quite well. The schematics are also well presented and have detailed PCB overlays. Probably the only negative is that some half-tone detail has been lost from the original manual as it has been scanned in simple B&W.
Perfect source for service manuals: fast and professional transaction; high quality, perfect readable and largely scaleable PDF; complete schemes, diagrams and spare part list. Tnx a lot, cu again!!!!
I got your link from a friend and I must say that I am really satisfied with your service. Specially this B&O manual I didn't find anywhere on the web... but you could deliver it :-) . You deliver very fast and the copy is of good quality. So your webpage is bookmarked. Thanks
This was the Sony CCU-500A Service manual I was looking for.
The price was reasonable.
The permission to download was quck.
I will use Owner-Manual.com for all my manual needs.
Excellent printing quality.
A complete and very usefull service manual with all details.
GREAT SERVICE AT VERY LOW PRICE!
Electronic clinical thermometer HF 365
Philips Domestic Appliances and Personal Care
Features : Memory Last measured temperature appears on the display after pressing the on/off button for at least 2 seconds. The buzzer sounds 3 times and the ÂºC symbol stops blinking. The thermometer may be immersed in disinfectant containing water for maximally one hour. The water temperature must be lower than 60 ÂºC. On display After 30 minutes Technical Data Power consumption Less than 60 micro-Watts, which means more than 3000 measurements with one battery. For this thermometer no spare parts will be available in the after sales period. When the product is defective, it has to be replaced. LCD display with 4.8 mm character height 32.00 ÂºC - 42.00 ÂºC Approximately 60 seconds Peak hold system Measurement is ready when the temperature does not change during 20 seconds. 0.1 ÂºC between 35.5 ÂºC and 42.0 ÂºC 0.2 ÂºC between 32.0 ÂºC and 35.5 ÂºC Button-cell 1.5 V type LR41 Sensor-tip: Stainless steel Housing parts and protective box:ABS p.t.o. p.t.o. p.t.o.
Measurement -readysignal Waterproof
Display Measuring range Measuring time Measuring system Measurement ready
Battery low indication Automatic switch - off
The thermometer meets the Medical Device Directive 93/42/EEC for medical appliances
Display informations Replacing the battery Cleaning instructions
Published by Philips Domestic Appliances and Personal Care
Printed in the Netherlands
Â© Copyright reserved
Subject to modification
4822 729 22062 98/11
PCS 99 803